26/02/2013 Trucks - Commercial Vehicles
Iotadental is supplying dental equipment to worldwide with free shipping. Not only the price is an obvious superiority, but also the top class quality and after-sale service. Here you can find variety dental products you want, and get them at surprising price. Dental Curing Light, dental intraoral camera, dental handpiece, dental lab equipment and anything you want. Come and join us, we are here waiting for you!
Specifications of Sun Autoclave Sterilizer 18L Vacuum Steam
* Rated Voltage: 220V±22V, 50Hz±1Hz, 110V±11V, 60Hz±1Hz
* Rated Power: 1800VA / 2000VA
* Sterilizing (Pressure/temp.): 0.9-1.3bar/120ºC, 2.0-2.3bar/120ºC (Both of above are Under Standard Atmospheric pressure)
* Fuse-Melting pipe: T15A / T30A
* Capacity of water tank: 2L
* Water Consumption volume per recycling round: 0.3L
* Operating Temp: 0-40ºC
* Inside dimensions: 245*355mm
* Outside dimensions: 650*520*500 mm
Features of Sun Autoclave Sterilizer 18L Vacuum Steam
* The Autoclave is specially designed for the occasions in use of frequent sterilizations. It is equipped with the microprocessor for intelligent control and man-machine interface for convenient operation.
* With the distinctions in displaying parameter and condition actively, evaluating malfunction automatically, shutting off for self-protection automatically in case of excessive temperature and pressure, this type autoclave is reliable product in sterilizing.
* This product can be applied in sterilizing for the tooth device, operation device, Sanitary Material and other heat-resisting articles existing in the Dept. of Stomatology, Ophthalmology and lab.
* Preparation Before Operation
1. Before operation, please connect well to the power source.
2. Connect the power source hole under the back left side of the autoclave and the power socket by the attached power cable inside the product’s package so as to switch on the autoclave.
3. Push down the main power button at the right bottom corner of the front side, when the indicator light is on, it means that the autoclave get through with the shown on the screen.
4. The autoclave is not heated while under such initialization status.
* The product can be applied in sterilizing for the tooth device, operation device, Sanitary Matetial and other heat-resisting articles existing in teh Dept. of Stomatology, Ophthalmology and lab.
* The autoclave must be plced on the horizontal working base.
* Must use the distilled water for the purpose of lasting the working life.
* There shall be no clog to the air cooling window on the outer surface.
* The object apparatus to be sterilized shall be palced on teh device plate with enough clearnace retained in favor of the air circulation inside the sterilizing room.
* Empty the water inside the cooling water collecting tank frequently. Usually, the cooling water collecting tank shall be emptied once the water storing tank emptied.
* Must revolve teh door handle to the teminal side while working.
* Don't open the door of autoclave when seeing the "0" displayed on teh pressure indicator.
* Not be too closed to the door of the autoclave in case of scald when open the door.
* Shut off the power before discharging/installing the sealing ring, and the work shall be carried out after further sufficient cooling in case of scald.
* Don't drag or drop the autoclave during relocation period of the autoclave.
* The protecting ground shall be reliable.
* Equipment must be used away from teh magnetic field.
Welcome to iotadental.co.uk where you can find more dental instruments.
Features of Sun® Medical Binocular 3.5 X BRILLENLUPE Kopflupe Lupenbrille Loupe 1. High quality with superior optics 2. Light, portable and easy to op...